Clinical Bibliography: Cerebral Oximetry for Beach Chair Procedures View
Case Study: Partial Anomalous Pulmonary Venous Return (PAPVR) and Bicuspid Aortic Valve, Nonin Medical’s SenSmartTM Universal Oximetry View
Understanding Accuracy, Bias, and Variability in SpO2 Readings View
Accuracy and Superior Performance of PureSAT and PureLight Oximetry Technologies View
Case Study: Using SenSmart Regional Oximetry to Aid in Transfusion Decisions View
Multi-Modality Monitoring in Carotid Surgery: The Importance of Reliable Readings View
Continuous Cerebral Oximetry Patient Monitoring During Post-Operative Transport Identifies Desaturation View
Accurate Oximetry Monitoring – See the Technology Difference View
Cerebral Oxygen Desaturation Predicts Cognitive Decline and Longer Hospital Stay After Cardiac Surgery View
SenSmart 8006 Hybrid Sensor Specifications
rSO2 Accuracy (Arms*): 50% to 100% rSO2
rSO2 accuracy testing was conducted during induced hypoxia studies on male and female, over 18 years of age, healthy, light- to dark-skinned subjects in an independent research laboratory. The measured regional hemoglobin saturation value (rSO2) of the sensors was compared to arterial/venous hemoglobin oxygen (SavO2) value, determined from venous and arterial blood samples. The model used for blood in the brain was 70% venous and 30% arterial, which is applicable under normocapnic conditions. The venous blood was drawn from the right jugular bulb. The accuracy of the sensors in comparison to the blood gas analyzer samples measured over the rSO2 range of 45–100%. Accuracy data was calculated using the root-meansquared (Arms value) for all subjects, per ISO 80601-2-61:2011, Medical Electrical Equipment—Particular requirements for basic safety and essential performance of pulse oximeter equipment.
Inter/Intra Sensor Repeatability Accuracy: ±2 digits (Arms*)
Operating: -5 °C to 40 °C (23 °F to 104 °F)
Storage/Transportation: -30 °C to 70 °C (-22 °F to 158 °F)
Device temperature will not exceed 41 °C as measured during a controlled environment test.
Operating: 10 % to 90 % non-condensing
Storage/Transportation: 10 % to 95 % non-condensing
* ±1 Arms encompasses 68% of the population at zero bias.
Magnetic Field Strength (8006PI):
Less than 1 gauss (0.1T) at a distance of 12cm.
Measurement Wavelengths and Output Power**
730 nanometers @ 3.0 mW maximum average power
760 nanometers @ 4.5 mW maximum average power
810 nanometers @ 3.2 mW maximum average power
880 nanometers @ 4.5 mW maximum average power
** This information is especially useful for clinicians performing photodynamic therapy.
This product complies with ISO 10993-1.
Nonin Medical Inc
13700 1st Ave N,
Plymouth, MN 55441
Toll Free (USA and Canada)
Nonin Medical B.V.
Doctor Paul Janssenweg 150
5026 RH Tilburg