8000FI and 8000FC Fiber Optic Sensors Specifications
SpO2 Accuracy1, 2, 3:
SpO2 Low Perfusion Accuracy: 70% to 100% ±2 digits (Arms*)1
Pulse Rate Accuracy: 20 to 250 BPM ±3 digits (Arms*)1
Pulse Rate Low Perfusion Accuracy: 40 to 240 BPM ±3 digits (Arms*)1
Temperature: 4, 5
Operating: 0 °C to 40 °C (32 °F to 104 °F)
Storage/Transportation: -30 °C to 50 °C (-22 °F to 122 °F)
Humidity: 4, 5
Operating: 10% to 95% non-condensing
Storage/Transportation: 10% to 95% non-condensing
* ±1 Arms encompasses 68% of the population at zero bias.
1 Additional accuracy and performance information can be found in the sensor accuracy document on the operator’s manual CD.
2 Accuracy testing was performed under no-motion conditions.
3 Accuracy specifications based on Nonin’s PureSAT® SpO2 technology and PureLight® sensor technology.
4 For combined oximeter/sensor specifications, refer to the applicable oximetry system’s operator’s manual.
5 Range as tested with Nonin’s PureSAT SpO2 technology.
Measurement Wavelengths and Output Power**
Red: 660 nanometers @ 0.8 mW maximum average
Infrared: 910 nanometers @ 1.2 mW maximum average
** This information is especially useful for clinicians performing photodynamic therapy.
This product complies with ISO 10993-1.
Not made with natural rubber latex.
The 8000FC/8000FI is warranted for 90 days from delivery.
Nonin reserves the right to make changes and improvements to these instructions and the product it describes at anytime, without notice or obligation.
Nonin Medical Inc
13700 1st Ave N,
Plymouth, MN 55441
Toll Free (USA and Canada)
Nonin Medical B.V.
Doctor Paul Janssenweg 150
5026 RH Tilburg